Fingertip Oxygen Monitor SPO2 PR 50D
Type:Visual Acuity Examination Apparatus
Place of Origin:Guangdong, China (Mainland)
Item:Digital finger pulse oximeter
display:1.3'' double color OLED
Display Format:64k color OLED
Pul Ratio se:30-255 BPM
Display Mode:Four Directions
Pleth Wave: OLED Graphic Display
fingertip pulse oximeter
FDA, CE and ISO certificate approve
Color LCD display with four directions
Small size and easy-ope
Finger Pulse Oximeter Fingertip Oxygen Monitor SPO2 PR Pulsoximeter
Principle of the 50D Pulse Oximeter is as follows: Photoelectric Oxyhemoglobin Inspection
Technology is adopted in accordance with Capacity Pulse Scanning & Recording Technology,the
Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through
finger.The product is suitable for being used in family, hospital, oxygen bar, community healthcare,
physical care in sports (It can be used before or after doing sports, and it is not recommended to
use the device during the process of having sport) and etc.
1. Integrated with SpO2 probe and processing display module
2. Small in volume,light in weight and convenient in carrying
3. Operation of the product is simple ,low power consumption
4. SpO2 value display
5. Pulse rate value display, bar graph display
6. Pulse waveform display
7. The display mode can be changed
8. Screen brightness can be changed
9. Low-voltage indication: low-voltage indicator appears before working abnormally
which is due to low-voltage.
Automatically power off function: when the device is under the state of measuring
interface . it will automatically power off within 5 seconds if the finger falls out of probe.
* Display format can be saved after power off
1. Display Mode:0.96" Dual-color OLED display(blue and yellow)
2. Screen Resolution:128*64
3. SpO2 Measuring Range:0%~100%, (the resolution is 1%).
4. Accuracy:70%~100%:±2% ,Below 70% unspecified.
5. PR Measuring Range:30bpm~250bpm, (the resolution is 1bpm)
6. Accuracy:±2bpm or ±2% (select larger)
7. Measurement Performance in Weak Filling Condition:SpO2 and pulse rate
can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ±4%,
pulse rate error is ±2 bpm or ±2% (select larger).
8. Resistance to surrounding light: The deviation between the value measured in the
condition of man-made light or indoor natural light and that of darkroom is less than ±1%
9. Power Consumption:less than 30mA
10. Voltage: DC 2.6V~3.6V
11. Power Supply:1.5V (AAA size) alkaline batteries × 2
12. Battery working hour: Theoretical number is 32 hours.
13. Safety Type:Interior Battery,BF Type
Sell in standard
1. a hanging rope
2. a user manual
1. Dimension:57(L) × 31(W) × 32(H) mm
2. Weight:About 50g (with the batteries)
1. 1 x Fingertip Oximeter
2. 1 x Lanyard
3. 1 x English user’s manual
4. 1 x Pouch.
We are Patient Monitor products,Patient Monitor medical,Patient Monitor technology,Patient Monitor news,Patient Monitor hospitals,Patient Monitor doctors,Patient Monitor articles,Patient Monitor vital signs,Patient Monitor blood pressure,Patient Monitor home,ECG & EKG watches,ECG & EKG heart,ECG & EKG cheat sheets,ECG & EKG nursing students,ECG & EKG cardiology,ECG & EKG awesome,ECG & EKG heartbeat,ECG & EKG monitor,homemade Incubator,Incubator diy,business Incubator,Incubator tray,manufacturers in China.Unified Wholesale price.Welcome to inquiry and OEM.